# ⓘ Plethysmograph. A plethysmograph is an instrument for measuring changes in volume within an organ or whole body. The word is derived from the Greek plethysmos, ..

## ⓘ Plethysmograph

A plethysmograph is an instrument for measuring changes in volume within an organ or whole body. The word is derived from the Greek "plethysmos", and "graphos".

### 1.1. Organs studied Lungs

Pulmonary plethysmographs are commonly used to measure the functional residual capacity FRC of the lungs - the volume in the lungs when the muscles of respiration are relaxed - and total lung capacity.

In a traditional plethysmograph, the test subject is placed inside a sealed chamber the size of a small telephone booth with a single mouthpiece. At the end of normal expiration, the mouthpiece is closed. The patient is then asked to make an inspiratory effort. As the patient tries to inhale a maneuver which looks and feels like panting, the lungs expand, decreasing pressure within the lungs and increasing lung volume. This, in turn, increases the pressure within the box since it is a closed system and the volume of the box compartment has decreased to accommodate the new volume of the subject.

### 1.2. Organs studied Methodological approach

Boyles Law is used to calculate the unknown volume within the lungs. First, the change in volume of the chest is computed. The initial pressure of the box times its volume is considered equal to the known pressure after expansion times the unknown new volume. Once the new volume is found, the original volume minus the new volume is the change in volume in the box and also the change in volume in the chest. With this information, Boyles Law is used again to determine the original volume of gas in the chest: the initial volume unknown times the initial pressure is equal to the final volume times the final pressure. Starting from this principle, it can be shown that the functional residual capacity is a function of the changes in volume and pressures as follows:

F R C = K V Δ V b o x Δ P m o u t h, K V ≈ P B a r o m e t r i c {\displaystyle FRC=K_{V}{\frac {\Delta V_{box}}{\Delta P_{mouth}}},K_{V}\approx P_{Barometric}}

The difference between full and empty lungs can be used to assess diseases and airway passage restrictions. An obstructive disease will show increased FRC because some airways do not empty normally, while a restrictive disease will show decreased FRC. Body plethysmography is particularly appropriate for patients who have air spaces which do not communicate with the bronchial tree; in such patients helium dilution would give an incorrectly low reading.

Another important parameter, which can be calculated with a body plethysmograph is the airway resistance. During inhalation the chest expands, which increases the pressure within the box. While observing the so-called resistance loop cabin pressure and flow, diseases can easily be recognized. If the resistance loop becomes planar, this shows a bad compliance of the lung. A COPD, for instance, can easily be discovered because of the unique shape of the corresponding resistance loop.

### 1.3. Organs studied Limbs

Some plethysmograph devices are attached to arms, legs or other extremities and used to determine circulatory capacity. In water plethysmography an extremity, e.g. an arm, is enclosed in a water-filled chamber where volume changes can be detected. Air plethysmography uses a similar principle but based on an air-filled long cuff, which is more convenient but less accurate. Another practical device is mercury-filled strain gauges used to continuously measure circumference of the extremity, e.g. at mid calf. Impedance plethysmography is a non-invasive method used to detect venous thrombosis in these areas of the body.

### 1.4. Organs studied Genitals

Another common type of plethysmograph is the penile plethysmograph. This device is used to measure changes in blood flow in the penis. Although some researchers use this device to assess sexual arousal and sexual orientation, "Courts that have considered penile plethysmography generally rule that the technique is not sufficiently reliable for use in court." An approximate female equivalent to penile plethysmography is vaginal photoplethysmography, which optically measures blood flow in the vagina.

### 2.1. Use in preclinical research Respiratory parameters from conscious freely moving animals: whole-body plethysmography

Whole-body plethysmography is used to measure respiratory parameters in conscious unrestrained subjects, including quantification of bronchoconstriction.

The standard plethysmograph sizes are for the study of mice, rats and guinea pigs. On request, larger plethysmographs can also be manufactured for other animals, such as rabbits, dogs, pigs, or primates.

The plethysmograph has two chambers, each fitted with a pneumotachograph. The subject is placed in one of them subject chamber and the other remains empty reference chamber.

The pressure change is measured by a differential pressure transducer with one port exposed to the subject chamber and the other to the reference chamber.

### 2.2. Use in preclinical research Respiratory parameters from conscious restrained animals: double-chamber / head-out plethysmography

The double-chamber plethysmograph dcp measures respiratory parameters in a conscious restrained subject, including airway resistance and conductance. Different sizes of plethysmograph exist to study mice, rats or guinea pigs.

The head-out configuration is identical to the standard configuration described above except that there is no head chamber.

Of course the collar seal is still applied, so that the body chamber remains airtight. With only a thoracic signal, all parameters can be obtained except for specific airway resistance SRaw and specific airway conductance Sgaw.

### 2.3. Use in preclinical research Resistance/compliance from sedated animals

In anesthetized plethysmography, lung resistance and dynamic compliance are measured directly because the subject is anesthetized.

Depending on the level of sedation, the subject may be spontaneously breathing SB configuration or under mechanical ventilation MV configuration. A flow signal and a pressure signal are required to calculate compliance and resistance.

### 2.4. Use in preclinical research Cerebral blood flow

Cerebral venous blood flow has been recently studied trying to establish a connection between Chronic cerebrospinal venous insufficiency and multiple sclerosis. The small study is not big enough to establish a conclusion, but some association has been shown.

• inflatable cuff at the base of the penis. Corpora cavernosa nerve penile plethysmographs measure changes in response to inter - operative electric stimulation
• lifted from each and invite the listener to spot the difference. Plethysmograph which deals with the government tool which measures arousal in sex
• air - displacement plethysmographs were ever developed for common, everyday use. Later experimental air - displacement plethysmographs developed in the 1980s
• stimulation to the nerve and measure the erectile function with a penile plethysmograph This test aids the surgeon in identifying the nerves, which are difficult
• blood flow. Labial thermistor clip Clitoral photoplethysmograph Penile plethysmograph Chivers ML, Seto MC, Lalumiere ML, Laan E, Grimbos T 2010 Agreement
• cavernous nerves of penis or apply an electrical stimulation penile plethysmograph diagnostic test to verify the nerves. The bilateral approach attempts
• thromboembolism. In the 1980s Nicolaides developed the first calibrated air - plethysmograph that could measure leg volume during exercise accurately. The measurements
• measurement using the sphygmomanometer change in body volumes using plethysmograph audiometry, eye examination, and many others. Bioluminescence imaging
• the early 1990s. It was reported that he previously used the penile plethysmograph before developing the Abel Assessment. In the Abel Assessment, the subject
• instrument for assessing female sex offenders. Labial thermistor clip Penile plethysmograph Vaginal photoplethysmograph Gerritsen, J., van der Made, F., Bloemers
• discomfort with using the device. Clitoral photoplethysmograph Penile plethysmograph Vaginal photoplethysmograph Chivers, M. L., Seto, M. C., Lalumiere
• synthesizer Palm Products GmbH, an audio synthesizer company Penile plethysmograph a device to measure male arousal Push Proxy Gateway, a component of